The weighing sensor is installed, which occupies an important position in the whole measurement process. If the weighing sensor is installed correctly, it will undoubtedly add to the whole measurement process, is conducive to improve the measurement accuracy, but also can effectively extend the service life.
Weighing sensor is the most stable installation of CBE1-20K cutting machine weighing sensor
The weighing sensor is installed, which occupies an important position in the whole measurement process. If the weighing sensor is installed correctly, it will undoubtedly add to the whole measurement process, is conducive to improve the measurement accuracy, but also can effectively extend the service life. Therefore, the installation process of the weighing sensor is very important:
1 sensor mounting surface and the mounting base should be maintained at a horizontal, non skew.
2 the difference of the level between the mounting surface of the weighing system is less than 5mm, and if the system has low precision, it can also be relaxed. 5/1000.
3 sensor mounting surface should be smooth, bright and clean, can not be installed on the surface of plastic film, burr, sharp points, etc..
4 install the surface base to be firm and maintain a certain thickness.
5 the base area of the mounting surface should be larger than that of the weighing sensor.
6 installation and replacement of the sensor, the need to select the appropriate torque wrench, adjust to the torque required to tighten the sensor.
7 need to install the gasket sensor, you need to install the gasket on the bolt can be installed.
8 in the fastening bolts, the need to apply a little butter to prevent rust and easy to install bolts.
9 prohibited in multi sensor cable weighing sensor, weighing system, lengthen or cut a part of the sensor cable.
10 in a single weighing sensor weighing system, such as the installation conditions permit, preferably not lengthened or cut the sensor cable.
The 11 cable access terminal box, signal line of each sensor should be connected to the corresponding terminal position, no 2 or more than 2 cable signal lines are simultaneously connected to a terminal.
12 when installing the upper and lower connector, it must be vertical with the weighing sensor, without deflection.
13 vertical and horizontal spacing, in the installation, maintenance, in accordance with the provisions of the installation instructions in the installation of equipment adjustment.
14 sensor wiring, the wiring box part connection hole when not in use, be sealed by the plug is blocked, in order to avoid moisture into the gray, the fastening bolts on the junction box cover loose.